Operate BTS channels in a batch Some customers asked us how to operate BTS channels together, such as how to start up tests together and how to export data all together. In fact, it’s easy. Please refer to the following pictures. Step 1: In the main working area, you can use CTRl+A to select all […]
Change views in BTS software client By clicking the buttons in the following pictures, you can switch different views in Neware BTS software client.
Easiest way to get the information of Neware BTS middle machine When new customers try to connect Neware battery testers with their computer, the 1st question is how to tell the status of middle machine. Because the information of middle machine is very important, right? Middle machine is the bridge between testers and computers. If […]
How to migrate BTS database to a new computer For some customers, they’d like to keep all the BTS testing history data in their computers, but as time goes by, the data size becomes larger and larger, in this situation, they want to migrate the BTS database to a newer and better computer. How to […]
How to set up high frequency test script correctly? For some customer, they’d like to use BTS4000 for high frequency(such as 500ms), so they’d set up a testing script likes the following picture: But when looking at the testing result, the data is not correct, please refer to the following picture, the current is very small […]
Careful using 0.1s or 1s data sampling interval for battery life cycle tests For Neware BTS4000 client software, the default sampling time interval is 0.1s, which means the testing data will be recorded 10 times each second. For short time test, this default setting is good, cause more data more helpful for you to get more […]
How to find out the mid machine and tester ID Sometime when you have trouble in using Neware BTS, the technicans from Neware may ask you the mid machine or the tester device ID, do you know how to find out the mid machine and tester ID from Neware BTS software? Please follow the following step, […]